3-1 CHAPTER 3 QUANTITATIVE MEASUREMENTS LEARNING OBJECTIVES Upon completion of this chapter, you will be able to do the following: 1. Explain the purposes and benefits of performing quantitative measurements. 2. Identify the various methods of performing impedance measurements. 3. Identify the various methods of performing power measurements. 4. Identify the various methods of performing frequency measurements. INTRODUCTION TO QUANTITATIVE MEASUREMENTSYou have already studied the basics of performing electronics measurements and how to determine if a component is or is not functioning properly. This chapter will cover techniques used in measurements of specific impedance, frequency, and power. These measurements are extremely important to you in evaluating the performance of a piece of electronic equipment. IMPEDANCE MEASUREMENTS Impedance measurements are often used during routine test procedures. Impedance-measuring equipment, such as impedance bridges, are mainly used in determining the capacitance and inductance of component parts. However, the values of combined circuit constants also may be obtained and used in direct calculations of impedance. An impedance measurement effectively totals the inductive and capacitive reactance together with the resistance in a circuit. In addition, impedance measurements are useful in testing and analyzing antenna and transmission line performance and for determining the figure of merit (Q) of electrical parts and resonant circuits. Q meters are impedance-measuring instruments that determine the ratio of reactance to resistance of capacitors or inductors and resistors. Details of Q meters and impedance bridges as well as a number of other methods of measuring circuit impedance are described in the following paragraphs. Also discussed are methods of measuring the impedance of antennas and transmission lines. BRIDGE METHODSBridges are among the most accurate types of measuring devices used in the measurement of impedance. In addition, bridges are also used to measure dc resistance, capacitance, and inductance. Certain types of bridges are more suitable for measuring a specific characteristic, such as capacitance or inductance. Basic schematics for the various bridge circuits are shown in figure 3-1. The bridge circuits shown are similar in that they usually contain two branches in the measuring circuit, two branches in the comparing circuit, a detector circuit, and a power circuit, as shown in figure 3-2. The bridge shown in figure 3-2 is actually the dc Wheatstone bridge; however, the general principles of circuit operation for ac remain the same.

Integrated Publishing, Inc.

6230 Stone Rd, Unit QPort Richey, FL34668

Phone For Parts Inquiries: (727) 493-0744
Google +